In the entry for today, we wanted to provide you with some insight into the Torrey LPC-40L price computing scale. The scale does price computing functions and is NTEP approved, legal for trade. Now in full disclosure we heavily promoted this scale a few years back and we do still offer this scale to customers who wish to purchase it. However, we are not big fans of the brand.
The LPC-40L scale is a great deal and affordably priced. The scale is built very well and has a rugged enclosure. The display is visible to both the scale operator and the customer. The display has a bright green backlight which makes it very easy to read, however the display can be a little hard to read without the backlight. The power is provided by standard 110VAC power and/or the internal rechargeable battery. Generally speaking the LPC has been fairly reliable. Over the years we have sold several scales that were also the lowest priced in the marketplace and they were constantly needing parts and always received terrible reviews by customers. However, the LPC40L has been fairly popular with customers and typically if something does break on the scale, it is usually something that can be repaired or replaced fairly easily.
In our experiences, the LPC-40L typically has the following parts failures: keypads, battery, power issues. When we have one of these scales come in for repair, it is usually either a battery or keypad issue. Every so often it will be something to do with the power, either the AC adapter or the plug where the power adapter plugs in has come loose. Now the repair costs for something like that can range anywhere from around $25 to $100 including the labor. So if the repair is around $100 you have to decide if it’s worth repairing the scale or just spending a little more money and buying a new scale. The weighing platter is a little awkward and high compared to most similar scales. This is fairly common on all Torrey counter top scales. If you need to weigh items in a different unit other than pounds, it can be tricky to change the weight units. Continue reading